NeoECG is the latest tablet smart ECG machine from Lepu Medical. This ECG machine is divided into a 10.1-inch version T180 with a printing dock and a 7-inch version S120 with a portable Bluetooth printer, both machines adopt the Android operating system and are also equipped with high-resolution capacitive touch screens. lightweight and portable, which is completely satisfied the needs of families, clinics, medical institutions, hospitals, and other scenarios.
High Accuracy Rate: Tested by 50,000,000 training data and 1,000,000 independent measured data, the average accuracy rate of the AI-ECG platform reach 95.2%.
High Analysis Speed: Just take 1s for automatically resting ECG analysis. The time saved can reduce the overall time of clinical ECG analysis.
Comprehensive Diagnosis: Support 16 types of cardiac classification, and 104 types of ECG diagnostic classifications.
10.1'' high-resolution color touch screen, easy to operate. Portable design, compact in size.
Support simultaneous acquisition and display of 18-channel waveforms, as well as cardiac/heart rate detection.
Support wireless transmission of ECG data via WiFi and mobile network.
Can be powered by an external DC power supply, and a built-in rechargeable lithium battery.
Support auto, RR analysis, HRV, medicine test, and ECG event mode.
Provide 4 sampling modes: pre-sampling, real-time sampling, periodic sampling, and trigger sampling.
Input patient information via a virtual alphanumeric keyboard and barcode scanning.
Freeze the ECG waveform on the screen.
272.6mm x181.2mm x 25.8mm
(Length x Width x Height)
274mm x 288 mm x 111.2 mm
(Length x Width x Height)
|Weight||1.2 Kg (including main unit, battery, without accessories)||About 2.5 kg (including battery)|
|Built-in rechargeable lithium ion battery||Rated voltage: 7.4V||Rated voltage: 14.4V|
|Rated capacity: 5000 mAh||Rated capacity: 4400 mAh|
|Power supply time: When only the internal battery is used, the standby time is more than 72 hours, and the continuous working time is more than 12|
|hours under normal circumstances after the battery is fully charged.|
(collected every 5 minutes)
|Charging time: Charge at least 5 hours before using the battery for the first time; when the battery is turned off and charged, the exhausted battery will be charged to 90% in less than 4 hours and fully charged in less than 5 hours.|
|Screen||10.1-inch, color LCD touch screen resolution: 1280 × 800 pixels|
|AC power||Rated voltage: 100 - 240 v~|
|Rated frequency: 50/60Hz|
|Recording paper||Reel thermal recording paper, paper width: 216 mm|
|Paper driving rate||5 mm/s, 6.25 mm/s, 10 mm/s, 12.5 mm/s, 25 mm/s, and 50 mm/s. The error is not greater than ±3%|
|Number of waveforms||Record up to 18 waveforms at the same time|
|Print resolution||Vertical resolution: 8 dots/mm Horizontal resolution: 40 dots/mm (Paper driving rate 25 mm/s)|
|Calculation method||Peak value detection|
|Measuring range||30 bpm - 300 bpm|
|Measurement accuracy||±1 bpm or ±1%, whichever is the largest|
|Lead||9-lead, 12-lead, 15-lead or 18-lead simultaneous acquisition and analysis|
|Sampling rate||32000 dots/s|
|Common mode rejection ratio (CMRR)||≥140 dB (Power frequency notch is turned on) ≥120 dB (Powerfrequency notch is turned off)|
|Frequency response||Based on 10 Hz, (0.01Hz- 350Hz)+0.4dB-3.0 dB|
|Sensitivity||Auto, 2.5 mm/mv, 5 mm/mv, 10 mm/mv, 20 mm/mv, and 40 mm/mV. The error is not more than ±5%|
|Filter||AC filter: 50 Hz, 60 Hz, and off|
|EMGfilter: 25 Hz, 35Hz, 45 Hz, and off|
|Baseline drift filter: 0.01 Hz, 0.05 Hz, 0.32 Hz, and 0.67Hz|
|Low pass filter: 75 Hz, 100 Hz, 150 Hz, 300 Hz, and off|
|Time base||5 mm/s, 6.25 mm/s, 10 mm/s, 12.5 mm/s, 25 mm/s and 50 mm/s.|
|Selection and accuracy||The allowable time measurement error within the range of 0.2s- 2.0s does not exceed ±3%.|
|Input impedance||≥100 MΩ (10 Hz)|
|Input loop current||≤10 nA|
|Polarization voltage||±900 mV, the waveform amplitude change is ±5%|
|System noise||≤12.5 µV|
|Amplitude quantification||0.95 µV/LSB|
|Electrode polarization recovery time||<10 s|
|Pace pulse display||Display ECG signal for pacing pulses with an amplitude of±2 mV-±700 mv, a duration of 0.1 ms- 2.0 ms, a rise time of less than 100 µs, and a frequency of 100/min.|
|Minimum detection||20 µVp-p|
Thank you for your attention on LEPU